Correction of measurement error in a commercial multiple-breath washout device - Wyler, Florian; Oestreich, Marc-Alexander; Frauchiger, Bettina S.; Ramsey, Kathryn; Latzin, Philipp (medRxiv, 2021)
Rationale Nitrogen multiple-breath washout (N2MBW) is an established technique to assess functional residual capacity (FRC) and ventilation inhomogeneity in the lung. Accurate measurement of gas concentrations is essential for the appropriate calculation of clinical outcomes.Objectives We investigated the accuracy of oxygen and carbon dioxide measurements used for the indirect calculation of nitrogen concentration in a commercial MBW device (Exhalyzer D, Eco Medics AG, Duernten, Switzerland) and its impact on FRC and lung clearance index (LCI).Methods High precision calibration gas mixtures and mass spectrometry were used to evaluate sensor output. We assessed the impact of corrected signal processing on FRC and LCI in a dataset of healthy children and children with cystic fibrosis using custom analysis software.Results We found inadequate correction for the cross sensitivity of the oxygen and carbon dioxide sensors in the Exhalyzer D device. This results in an overestimation of expired nitrogen concentration, and consequently FRC and LCI outcomes. Breath-by-breath correction of this error reduced mean (SD) FRC by 8.9 (2.2)% and LCI by 11.9 (4.0)%. It also resulted in almost complete disappearance of the tissue nitrogen signal at the end of measurements.Conclusions Inadequate correction for cross sensitivity between the oxygen and carbon dioxide gas sensors of the Exhalyzer D device leads to an overestimation of FRC and LCI. Correction of this error is possible and could be applied by re-analysing the measurements breath-by-breath in an updated software version.Grants, Gifts, Equipment, Drugs Eco Medics AG (Duernten, Switzerland) provided a research version of their commercial software Spiroware 3.2.1 including insight on signal processing algorithms and helped with the acquisition of mass spectrometry measurements. This project was funded by the Swiss National Science Foundation, Grant Nr. 182719 (P. Latzin) and 168173 (K. Ramsey)Competing Interest StatementAll authors are in regular contact with manufacturers of multiple breath washout (MBW) devices including Ecomedics AG (Duernten, Switzerland). The authors contacted Ecomedics AG in the process of this study to obtain further information on current signal processing and algorithms for the computation of MBW outcomes and present results. There were no changes to the manuscript by the Ecomedics AG.The authors declare no conflict of interest. Prof. Latzin: personal fees from Vertex, Novartis, Roche, Polyphor, Vifor, Gilead, Schwabe, Zambon, Santhera, grants from Vertex, all outside this work. All other authors declare no conflicts of interest. Funding StatementThis project was funded by the Swiss National Science Foundation, Grant Nr. 182719 (P. Latzin) and 168173 (K. Ramsey).Author DeclarationsI confirm all relevant ethical guidelines have been followed, and any necessary IRB and/or ethics committee approvals have been obtained.YesThe details of the IRB/oversight body that provided approval or exemption for the research described are given below:The Ethics Committee of the Canton of Bern, Switzerland approved the study protocol (PB_2017-02139).All necessary patient/participant consent has been obtained and the appropriate institutional forms have been archived.YesI understand that all clinical trials and any other prospective interventional studies must be registered with an ICMJE-approved registry, such as ClinicalTrials.gov. I confirm that any such study reported in the manuscript has been registered and the trial registration ID is provided (note: if posting a prospective study registered retrospectively, please provide a statement in the trial ID field explaining why the study was not registered in advance).YesI have followed all appropriate research reporting guidelines and uploaded the relevant EQUATOR Network research reporting checklist(s) and other pertinent material as supplementary files, if applicable.YesDue to the nature of this research, participants of this study did not agree for their data to be shared publicly, so supporting data is not available.