Intraluminal pressure in central venous hemodialysis catheters during power injection of contrast media - White, Nicholas A; van der Molen, Aart J; Limpens, Ronald W A L; Maas, Jacinta J; van der Bogt, Koen E A; Horeman, Tim; Rotmans, Joris I (medRxiv, 2024)
Background Central venous catheters (CVCs) provide direct access to the central circulatory system, commonly used in hemodialysis and intensive care units for drug administration. Although uncertified for the procedure, CVCs are sometimes used for power injection of contrast medium (CM) during CT scans to avoid peripheral intravenous catheter placement. Previous studies suggest this practice is safe, but incidents have been reported. This study aims to measure intraluminal pressure during CM injection through CVCs and assess its impact on the luminal surface to guide responsible clinical use.Methods Strain gauges were applied to the exterior walls of four samples from three different types of unused CVCs. These gauges measured material deformation due to intraluminal pressure during CM injections at rates of 4.5 mL/s and 8 mL/s, each performed five times. Strain data were calibrated against known pressures in a static system. The CVCs were then either pressurized until bursting or subjected to microscopic analysis of their luminal surfaces.Results Intraluminal pressures measured (97-545 kPa or 14-79 PSI) were below the burst pressure (779-1248 Kpa or 113-181 PSI) in all instances. Strain regression analysis shows a statistically significant (p<0.05) trend over 10 injections in almost all CVCs tested, indicating material fatigue. Surface microscopy revealed surface micro-cracks from repeated injections, suggesting material damage.Conclusions The intraluminal pressures from power injections of CM are sufficiently low to prevent CVC bursting. While incidental use for CM injection appears safe, repeated use may cause material damageCompeting Interest StatementThe authors have declared no competing interest.Funding StatementNo external funding was received for this work.Author DeclarationsI confirm all relevant ethical guidelines have been followed, and any necessary IRB and/or ethics committee approvals have been obtained.YesThe details of the IRB/oversight body that provided approval or exemption for the research described are given below:N/AI confirm that all necessary patient/participant consent has been obtained and the appropriate institutional forms have been archived, and that any patient/participant/sample identifiers included were not known to anyone (e.g., hospital staff, patients or participants themselves) outside the research group so cannot be used to identify individuals.YesI understand that all clinical trials and any other prospective interventional studies must be registered with an ICMJE-approved registry, such as ClinicalTrials.gov. I confirm that any such study reported in the manuscript has been registered and the trial registration ID is provided (note: if posting a prospective study registered retrospectively, please provide a statement in the trial ID field explaining why the study was not registered in advance).YesI have followed all appropriate research reporting guidelines, such as any relevant EQUATOR Network research reporting checklist(s) and other pertinent material, if applicable.YesAll data referred to in the manuscript is available upon reasonable request.CMcontrast medium/mediaCVCcentral venous catheter